The Selective Radiation Meter SRM-3006 is always delivered with the latest firmware version installed. As the firmware is being constantly developed, it is a good idea to update the device on a regular basis.
Use the SRM-3006 Tools or SRM-3006 TS PC software to update the device. Simply follow the instruction in the SRM operating manual or in the on line help for SRM-3006 TS.
The latest firmware contains:
"Safety Evaluation" mode
"Spectrum Analysis" mode
"Level Meter" mode
"Scope" mode (option)
"UMTS P-CPICH Demodulation" mode (option)
"LTE Demodulation" mode (option)
The options are charged extra and can only be activated by a special code. Contact Air-Met for more information.
Please note the following when updating the firmware:
A firmware update can take up to 30 minutes. Prematurely canceling the update can cause data loss and render the instrument inoperable.
Finish all measurements.
Export all the data you wish to retain to a PC, as such data could be lost during an update. If this happens, the data cannot be restored.
Make sure that the power supply is uninterrupted during the update. Connect the AC adapter/charger to ensure this.
Only start the update when you have done the above.
Wait until the update completes successfully before you change any settings on the SRM.
Please note that you will need to set up a user account in order to download the firmware.