NBM-550: Firmware Version 2.2.2 Update
The NBM-550 Broadband Field Meter is always delivered with the latest firmware version installed.
As the firmware is being constantly developed, it is a good idea to update the device on a regular basis.
The new Firmware Version 2.2.2 eliminates an error that occurred when making measurements together with the EHP-50F and which caused result values for frequencies below 5 Hz not to be transferred to the NBM-TS software during data transfer. The correction only affects measurements in EHP-50F/Spectrum operating mode in the frequency ranges Span = 100 Hz and 200 Hz.